1Tehran University of Medical Sciences, P.O. Box 14155-6145, Tehran, I.R. IRAN
2Chemical Engineering Department, Houldsworth Building, University of Leeds, Leeds LS2 9JT, UK
3Applied Mathematical Studies, School of Mathematics, University of Leeds, Leeds LS2 9JT, UK
Personal samplers or in general blunt body samplers are widely used in occupational hygiene for collecting air contaminants in the work environments. This work is part of an ongoing research into the performance evaluation of personal samplers, particularly in terms of their aerodynamic properties. Velocity profiles have been measured around and within typical cylindrical sampling devices, placed inside of a wind tunnel. A two-component fibre optic laser Doppler anemometer (LDA) is employed to measure the flow field around as well as within the samplers under the test. A variety of conditions, i.e., suction flux, wind speed and orifice size, have been examined. Extensive tests have been carried out in order to verify the reproducibility and reliability of measurements. The results show that reproducibility of the measurements at upstream of samplers are within 1% of the mean velocity. However, analysis of extensive data revealed the presence of noticeable electrostatic effect within and around the sampling device. Attempts were made to eliminate the presence of the electrostatic effect by, for example, spraying the nonconductive sampler with an anti electrostatic liquid, and wire earthing of a sampler made from cooper, but unfortunately these measures were found to be fruitless in eliminating the electrostatic effects. It should nevertheless, be said that the observed velocity is found to be more stable for small inlet orifice size of 4 mm and high suction flux of 3 l/min and this may be because the charged particles have less time to interact with the field produced by the sampler.
 Vincent, JH., “Aerosol Sampling Science and Practice”, New York, John Wiley & Sons, pp. 178, 283-297, 328 (1989).
 Comite´ Europe´en de Normalisation (CEN), Workplace Atmospheres Size Fraction Definitions for Measurements of Airborne Particles in the Workplace (CEN standard EN 481), Brussels, Belgium (1993).
 International Organization for Standardization (ISO), Air Quality-Particle Size Fraction Definitions for Health-Related Sampling (ISO standard 7708), ISO, Geneva, Switzerland (1995).
 American Conference of Governmental Industrial Hygienists, Threshold Limit Values for Chemical Substances and Physical Agents and Biological Exposure Indices, Cincinnati, ACGIH, Ohio, p. 85 (2005).
 Aizenberg, V., Grinshpun, S. A, Willeke, K., Smith, J. and Baron, P.A, Performance characteristics of the button personal inhalable aerosol sampler, Am. Ind. Hyg. Assoc. J., 61 (3), 398 (2000).
 Baron, P. A., Chen, C. C., Hemenway, D. R., Shaughnessy, P. O., Non-uniform air flow in inlets: The effect on filter deposits in the fiber sampling cassettes, Am. Ind. Hyg. Assoc. J., 55, 722 (1994).
 Vincent, J. H., Aerosol science for industrial hygienists, Elsevier Science, UK, pp. 261-268 (1995).
 Baron, P. A., Deye, G. J., Electrostatic effects in asbestos sampling I: Experimental measurements, Am.Ind. Hyg. Assoc. J.,51, 51 (1990).
 Paik, SY, Vincent, JH, The orientation - averaged aspiration efficiency of IOM-like personal aerosol samplers mounted on bluff bodies, Ann. occup. Hyg., 48 (1), p. 3 (2004).
 Baron, PA., Deye, G J., Electrostatic effects in asbestos sampling II: Comparison of theory and experiment, Am.Ind. Hyg. Assoc. J.,51, 63 (1990).
 Polat, H., Polat, M., Chander, Hogg, R., The electrostatic charge on particles and its relation to agglomeration in air, Appl. Occup. Environ. Hyg. J., 11, 817 (1996).
 Pourmahabadian, M., Experimental study of wind tunnel performance by a two-component laser doppler anemometer, Iran Occupational HealthJournal, 1 (1), 48 (2005).
 National Institute of Occupational Safety and Health, NIOSH Manual of Analytical Methods, http://www. cdc.gov/niosh/nmam/
 Hall, D.J., Upton, S. L., Marsland, G.W. and Waters, R. A., Wind tunnel measurements of the collection efficiency of two PM-10 samplers: the Sierra- Anderson Model 321 A Hi-Volume sampler and EPA prototype Dichotomous sampler, Warren SpringLaboratory, Report NoLR699 (1988).